Maintenance knowledge of electromagnetic flowmete
Maintenance knowledge of electromagnetic flowmeter
The electromagnetic flowmeter converter has a self-diagnostic function. In addition to power and hardware circuit failures,faults that occur in general applications can be correctly given.
Electromagnetic flowmeters are common faults
Maintenance and maintenance mainly have the following aspects:
- Excitation alarm
When this phenomenon occurs, check whether the excitation wiring x and y are open, check whether the resistance value of the excitation coil is normal ,and the determine whether the converter has malfunction .
- The measured flow is not accurate
When this happens ,it should be checked whether the fluid is filled with the sensor measuring tube ,the signal line connection is normal ,check the sensor coefficient, sensor zero? whether the point is set normally by the sensor label or the factory checklist.
- Meter no display
When this fault occurs, first check whether the power supply is turned on ,check whether the power supply fuse is intact ,and check whether the power supply voltage meets the requirements, such as if all the above are normal ,the converter should be sent to the manufacturer for repair.
- Empty tube alarm
When this problem occurs ,measure whether the fluid fills the sensor measuring tube and check that the signal wiring is correct. Check if the flow sensor electrode is normal .use a wire to short-circuit the converter signal input terminals a, b and c .at this time ,if the ¡°empty pipe alarm ¡°prompts to cancel ,the converter is normal.
There may be a low conductivity of the measured fluid or an empty tube threshold and an empty tube range setting error.
KFKC INSTRUMENT CO.,LTD specializes in the production of electromagnetic flowmeter ,turbine flow meter ,vortex flowmeter ,throtting device series for 25 years,complete specifications, various forms, can meet your different needs in different occasions.